|Place of Origin:||China|
|Model Number:||VP-OJ Pro|
Payment & Shipping Terms:
|Minimum Order Quantity:||1 set|
|Packaging Details:||85*42*46cm 20KGS|
|Delivery Time:||with 7 days|
|Payment Terms:||L/C, Western Union, MoneyGram, T/T|
|Feature:||Blood Vessels Removal, Skin Tightening, Skin Rejuvenatio||Power:||320W|
|Product Name::||Multifunction Beauty Machine||Application::||For Commercial & Home Use|
Peeling 6 In 1 Hydra Dermabrasion Machine,
Whitening 6 In 1 Hydra Dermabrasion Machine,
Dermabrasion hydrafacial 6 in 1 machine
6 in 1 Deep Clean Blackhead Removal Whitening Peeling Ance Removal Hydra Microdermabraision Beauty
|Technology||Non-invasive high pressure|
|Screen||10.0 inch 1024*600HD|
|Ultrasound||Frequency 3M capacitor 5600±15%PF|
|Hydrogen and oxygen cleaning handle||Pressure≈92KPa|
1. Non-invasive Water Mesotherapy Injection
Non-invaisve high-prssure 800KPA pressure injection, will produce high oxygen to quickly direct the nutritional products directly to the dermis layer, so that nutrients are easily absorbed and stimulate protein regeneration.
2. Plasma Sterilization
The use of high-frequency voltage electrolyzes oxygen in the air to produce ozone. It can kill bacteria, fungi and mites in a short time, and inhibit inlammation, and the effect of removing acne is obvious.
3. Hydrogen and Oxygen Cleaning
Up to 92KPa negative pressure suction, combined with hot and cold adjustable modes. It can instantly clean the hair follicles with the concentration of 1480ppb of hydroxide ionized water in the form of vortex, increasing skin elasticity.
4. Rolling Radio Frequency
The rolling ball simulates the kneading action, and the bipolar RF radio frequency Y shape is more suitable for stimulating the new reorganization of subcutaneous collagen. Aiming at sagging eys bags, eye corner wrinkles, and chin contour, from the inside to the outside, from top to bottom, the contour is tightened.
5. Cold Calm
Physical cooling, the principle of thermal expansion and contraction, the temperature is 0-5 degrees Celsius, it calms the skin and shrinks the capillaries, makes the pores tighten the skin, eliminates redness and swelling, and beautifies the skin.
6. Ultrasonic Head
Ultrasound activates skin cells at a frequency of 1 to 3 million vibrations per minute, promotes nourishment and absorption of cells, and makes the absorption rate of cells reach more than 90%.
Nursing standard operating procedures:
1)Choose care items: According to the customer's skin conditions and needs, choose the right
2)Customer file registration: fill in the customer file information in detail after treatment and
print it out, take pictures before treatment,
3)Care site planning: A management plan with prominent key points should be formulated
according to the specific conditions of the customer's skin.
4)Introduce the care plan: Introduce the care plan to the customer in detail, so that the customer
fully understands all the skin reactions before and after care and the matters that should be
paid attention to after care.
(Preliminary preparation: Prepare all the items needed for the operation before the operation: a
transparent cup, a container for the cleaning head, a brush, a set of original solution, masks, gloves,
a bottle of physiological saline, 75% alcohol, blackhead export Liquid, pore shrinking liquid, cotton
pad, toner (or growth factor), essence, lotion or cream moisturizing mask, non-woven fabric,
hyaluronic acid liquid.)
5)Clean the care area: thoroughly clean the skin and keep the skin dry; disinfect the hands of the
beautician and the bald head of the instrument to ensure the sterility in the operation.
(Note: Before treatment, all metal products such as rings, watches, necklaces, etc. worn by customers
should be removed)
6)Connect and check the instrument: Connect the instrument power and various accessories, the LCD
screen on the panel is off. After checking, it will enter the working state.
1) Turn on the instrument; turn on the power switch of the instrument, the display of the instrument
will light up, and the circuit of the system will immediately enter the self-checking state. In this case,
the operator should let the instrument run for 2-3 minutes before proceeding to the next operation.
2) Warm-up: Warm up the instrument for three minutes, do not use the instrument during warm-up.
3) Use with the product used.
Enter Your Message